## SINEAX M 561 Power & Current Transducer

SINEAX M 561 is a programmable transducer with a RS 232 C interface.

M 561 supervises 1 variable
(input) which is available on an analog output signal.

Input and
output are electrically isolated.

Rated current: 1..6A.
Rated voltage: 57,7...400V (P-N) rsp. 100...693V (L-L)

Universal analog outputs (programmable).
High accuracy: class 0.5

Sine, Rectangular or Distorted Wave Forms

One Analog Outputs

Measures any Three: P, Q or S, I ~,

U ~ (RMS), cos, sin j, LF, PF, QF, f

Data Logging Function

RS 232 PC Programming Interface

24-60 VAC/VDC or 85-230 VAC/VDC Powered

Input: 0-0.8 A to 0-1.2 A or 0-4 A to 0-6 A 57-400 V

phase-to-neutral 100-693 V phase-to-phase 50/60 Hz

Output:0-20 mA, 4-20 mA 0-1 V to 0-10 V 2-10 V

**SINEAX M 561 programmable transducer **

Model |
Description |
---|---|

SINEAX M561 |
Programmable Multi-transducer, 1 analog output, RS-232 |

## SINEAX M 562 Power & Current Transducer

M 562 measure 2
3 variables simultaneously and generate 2 electrically
isolated analog output signals.

The transducer is also equipped with an RS 232 serial interface to which a PC with the corresponding
software can be
connected for programming or accessing and executing useful
ancillary functions.

Rated current: 1..6A.
Rated voltage: 57,7...400V (P-N) rsp. 100...693V (L-L)

Universal analog outputs (programmable).
High accuracy: class 0.5

Sine, Rectangular or Distorted Wave Forms

3 Analog Outputs

Measures any Three: P, Q or S, I ~,

U ~ (RMS), cos, sin j, LF, PF, QF, f

Data Logging Function

RS 232 PC Programming Interface

24-60 VAC/VDC or 85-230 VAC/VDC Powered

Input: 0-0.8 A to 0-1.2 A or 0-4 A to 0-6 A 57-400 V

phase-to-neutral 100-693 V phase-to-phase 50/60 Hz

Output:0-20 mA, 4-20 mA 0-1 V to 0-10 V 2-10 V

**SINEAX M 562 power current transducer**

Model |
Description |
---|---|

SINEAX M562 |
Multi-transducer, CT inputs, 2 analog outputs, RS-232 |

## SINEAX M 563 Power & Current Transducer

For simultaneously acquiring 3 freely selectable measurands of an electrical power system.

Three analog outputs, RS232.

Programmable application (type of electrical system)

Rated current: 1..6A.
Rated voltage: 57,7...400V (P-N) rsp. 100...693V (L-L)

Universal analog outputs (programmable).
High accuracy: class 0.5

Sine, Rectangular or Distorted Wave Forms

3 Analog Outputs

Measures any Three: P, Q or S, I ~,

U ~ (RMS), cos, sin j, LF, PF, QF, f

Data Logging Function

RS 232 PC Programming Interface

24-60 VAC/VDC or 85-230 VAC/VDC Powered

Input: 0-0.8 A to 0-1.2 A or 0-4 A to 0-6 A 57-400 V

phase-to-neutral 100-693 V phase-to-phase 50/60 Hz

Output:0-20 mA, 4-20 mA 0-1 V to 0-10 V 2-10 V

**M563 3 analog outputs transducer**

Model |
Description |
---|---|

SINEAX M563 |
Multi-transducer, CT inputs, 3 analog outputs, RS-232 |

## SINEAX P 530 Active Power & Q 531 Reactive Power Transducer

Measuring transducer for the conversion of active power in single phase alternating current or
3-phase systems with balanced or unbalanced load.

A load-independent direct current or direct voltage is used as an output signal, which is proportional to
the measured active power value.

Measuring method: TDM method

Measurement inputs: sinusoidal nominal input current 1...6 A and nominal input voltage 100...690 V(L-L)

3-Phase with Balanced or Unbalanced Load

Sinusoidal Input Available as Measuring Source

Powered to 230 V 24-60 VAC/VDC or

85-230 VAC/VDC Powered

Input:0-0.8 A to 0-1.2 A or 0-4 A to 0-6 A

100-690 V 50/60 Hz

Output:0-1 mA to 0-20mA 1-5 mA to 4-20 mA

1 mA to 20 mA 0-1 V to 2-10 V, to 10 V

**Active Power Transducer**

Model |
Description |
---|---|

SINEAX P530 |
Active Power Transducer |

SINEAX Q531 |
Reactive Power Transducer |